HBLT-Wire can now be configured without using a PC!
We regularly upgrade the PC-based software used for setting up the sensors.The upgrades are based on input from our customers as well as on new features that we build into the sensors.We have now taken the steps to allow the configuration of the most sold type, HBLT-Wire, to be carried out without a PC.The setup is carried out using a small button on the front of the sensor.The three-figure display sends information back to the person carrying out the configuration using simple codes.
It is possible to configure the type of refrigerant, the length of the wire, and the size of the standpipe.
HBLT-Wire can be configured for the following applications:
|Parmeters to configure||Selection possibilities|
|Refrigerant:||R718, R717, R507, R410, R407, R404, R134, R123ZE, R22|
|Wire length:||300-4000 cm with 1 cm gap|
|Standpipe:||DN100, DN80, DN65, DN50, DN40, DN32, DN25|
HBLT-Wire can also be installed in the tank, i.e. without the use of a standpipe.Here, you just choose standpipe DN100 in the sensor configuration.
If HBLT-Wire is set up using the PC-based software, it is now also possible to assign the sensor offset values for the top and bottom.For example, with this feature, the sensor can be configured specifically to transmit a 100% (20 mA) measurement signal 200 mm from the sensor’s top and 100 mm from the sensor’s bottom (end).
Offset at the top can be relevant if the standpipe is placed higher at the top of the tank. Offset on the bottom can be relevant if oil accumulates in the standpipe due to inadequate drainage conditions in the pipe.